Payment & Shipping Terms:
|Brand:||OTS||Key Words:||Charpy And Izod Impact Testing Machine|
|Angle Measuring Range::||0-360°||Angular Resolution::||≤0.06°|
|The Host Weight (KG):||520||Main Motor::||550W|
|The Impact Blade Thickness::||16mm||Specimen Holder Support Surface Angle::||11°|
Charpy And Izod Impact Testing Equipment,
550W Impact Testing Equipment,
550W Charpy And Izod Impact Testing
Charpy And Izod Impact Testing Machine Angle Measuring Range : 0-360°
XB-OTS-W series Computer Screen Display Pendulum Impact Testing Machine is mainly used to determine the anti-impact capability of metal materials under dynamic load. Carry out the functions of zero clearing and automatic return, capturing the value of lost impact energy and pendulum cycle by means of setting up with computer program, and the results can be monitored, stored and printed out. Control box or computer program control is alternative operating method. XB-OTS-W series impact testing machine is adopted by many institutes and high-tech enterprises
Usage And Characteristics
The equipment is used for the determination of the impact resistance performance of metal materials under dynamic loads, in order to determine the nature of the material under dynamic load .The use difference between the potential energy before impact and the remaining energy after the impact displayed in the dial on the way, to get absorbed power sample. With one 250J pendulum hammer, with the cross-section of the sample of (10 * 10)mm.
Operation with semiautomatic control, convenient operation, high work efficiency, use of the surplus energy can be automatically raise the pendulum, can embody its superiority during the continuous impact test.
Main Technical Specifications:
|Impact energy (J)||250J|
|Dial scale range and division value:||The energy range: 0-250J
Each small lattice indexing values: 2.5J
|The pendulum torque:||0-250J M=133.9746N.m|
|Impact velocity (m/s)||5.4|
|Standard span (mm)||40|
|Distance between pendulum rotating center and specimen center (mm)||800|
|Round angle of jaws (mm)||R1.0-1.5 (1mm is special order)|
|Round angle of striking edge (mm)||R2.0-2.5 (8mm is special order)|
|The sample size: (mm)||10 x 10 (7.5 or 5) * 55|
|Overall dimensions (length * width * height)||2230×640×1480 mm|
|The impact of the blade angle:||30°|
|Specimen holder support surface angle:||11°|
|The impact blade thickness:||16mm|
|Angle measuring range:||0-360°|
|The host Weight (KG)||520|
|Power supply:||three phase four wire 380V 50Hz 700W|
(1) within the room temperature of 20°C±5°C;
(2) relative humidity is not more than 85%;
(3) the surrounding non corrosive medium, no vibration, no strong
electromagnetic interference environment;
(4) mounted on the concrete foundation thickness not less than 200mm or fixed on foundation more than 1400Kg;
GB/T 3808-2002 "Verification of pendulum-type impact testing machines";
J JG145-2007 "pendulum-type impact testing machine".
GB/T 229-2007 "Metallic materials-Charpy notch impact test";
ASTM E23-02a "Standard Test Methods for Notched Bar Impact Testing of Metallic Materials"
(can meet the standard while equip with American standard pendulum);
ISO 148.1-2009 "Metallic materials - Charpy pendulum impact test Part 1:Test method"
ISO 148-2:2008 "Metallic materials - Charpy pendulum impact test -Part 2: Verification of test machines".
ISO 148.3-2008 "Metallic materials - Charpy pendulum impact test Part 3:Preparation and
characterization of Charpy V-notch test pieces for indirect verification of pendulum impact machines"
ISO R83 "steel Charpy (CHARPY) test (U notch impact test of metallic materials)"
Contact Person: Andy Xiao