Payment & Shipping Terms:
|Name:||PVC Floor Casters Abrasion Durability Machine Floor Coverings Abrasion Tester||Test Function:||Caster-Resistant Flooring Effect Test / Resistant|
|Warranty:||1 Year||Sample Table:||800mm|
|Size Of Sample:||750mm||Test Speed:||Rotary Table (20 + 0 / -1) R / Min, Caster Assemble|
|Standard:||BS En-425 En-985 ASTM D6962-17 & ISO 4918||Origin:||China|
ASTM D6962-17 Abrasion Tester,
Caster Resistant PVC Floor Abrasion Tester,
ISO 4918 abrasion testing machine
PVC Floor Casters Abrasion Durability Machine Floor Coverings Abrasion Tester
Casters floor testing machine CE-MD certificate.pd...Casters floor testing machine CE-MD certificate.pd...
Caster-resistant floor friction tester, also known as floor casters abrasion durability testing machine, floor casters friction durability testing machine, also known as castor-resistant floor effect tester, etc., this test is widely used to test the wear resistance of the floor caster by certain Specifications casters in a certain speed, a certain load, friction floor samples under certain circumstances trajectory, automatically shut down after the completion of the cumulative number of trials to observe the change of the sample, the sample assess wear, with a total test cycle counting, cycle times It can be set, holding pressure time can be set and other functions.
BS EN 425-2002 (R2012): Resilient and Laminate Floor Coverings - Castor Chair Test Resilient floor coverings - Determination of chair casters effects.
BS EN 985-2001 Textile floor coverings -. Castor chair test textile floor casters wheelchair test
ISO 4918-2009 Resilient, textile and laminate floor coverings - Castor chair test stratified elastic textile floor coverings - foot wheelchair test
ISO 10581-2011 Resilient floor coverings - Homogeneous poly (vinyl chloride) floor covering - Specification Elastic Carpet - Homogeneous PVC carpet - Specification
ANSI-NALFA-LF-01-2008.3.6 / ANSI-NALFA-LF-01-2011 Castor Chair Resistance-- North American Laminate Flooring Association chair casters anti-abrasion test
|Test function||Caster-resistant flooring effect test / Resistant floor caster friction durability test|
|Size of Sample||¢750mm|
|Test Speed||Rotary table (20 + 0 / -1) r / min, caster assembly (50 ± 1) r / min|
|Timer||(0 ~ 999999) times ( the standard of 25,000 times can be set)|
|Run Time Control||After the rotation of the (180 ± 10) s, and maintain the platform stops (5 ± 2) s|
|Test Method||continuous cycle test|
|Power Supply||1 phase 220 V 50hz|
One set of standard load
One set of floor casters
One set of Power line,operation manual
Test Operation Specification
1. Prepare test samples
Take the test floor and check its function and appearance in advance;
The floor is adhered to the test turntable with double-sided adhesive, and no looseness is
2. Confirmation before test
Equipment power supply connection, ensure no foreign object interference, ensure no potential
safety hazard to operation;
The test sample and the required fittings are complete and in good condition.
3. Operation procedure
(1) Clean the surface of the castor and dry with a cotton pad or paper dipped in ethanol. Place the
test support platform on the test platform to ensure that the hole on the specimen holder is bolted
to the test platform. Slow down caster to caster contact test piece. Put the vacuum cleaner as close
as possible to the test piece, but do not touch. Start the vacuum cleaner.
(2) Testing requires a product with a substrate. If the substrate is not specified, the polyethylene
foam with a density of (35 + 5) kg/m3 shall be used as a substrate with (2 + 0.5) mm thickness.
Install the sample on the test support platform with double adhesive tape or adhesive tape to make
sure the double adhesive tape or the adhesive tape is covered with the whole specimen support
platform. Make sure there is no gap between the specimen and the top to prevent movement or
lifting of the specimen under loading conditions.
(3) Adjust the counter to the test set of test cycles and set the counter 25000 rpm.
(4) When the sample and parameter settings are ready, press the ALM key to execute the entire
(5) After the 25000 turn of the machine is stopped, each piece of the specimen is checked, and the
specimen after running 25000 rpm is evaluated. Before evaluation, 25000 RPM specimens shall be
maintained at least 24h, and the surface of the test specimen shall be upward or free suspension.
Attention: The test line of experimental equipment has been set at the factory speed, the line speed
to meet the requirements of the standard, no special circumstances that do not modify (change
mode: in the corresponding inverter electric control box adjusting the parameters can be
automatically calculated by changing the test line corresponding to the speed).
Contact Person: Elina Xu