Payment & Shipping Terms:
|Key Words:||Puncture Resistance Tester||Outside Dimension (L*W*H):||800*500*750 Mm|
|Impale Cone: Height::||1||Weight:||140kg|
|Sample:||175x175mm, 300x300mm||Unit ( Optional)::||Kg-cm, Joule, Or Designated|
|Brand:||OTS||Pointer Friction:||Pointer Friction|
Digital Puncture Resistance Tester,
Puncture Resistance Tester Computer Controlled,
CE Puncture Resistance Tester
Puncture Resistance Tester / Micro Computer-Controlled Technology
Digital type cardboard Tester is the dedicated equipment for corrugated cardboard.
Its parameter conforms to the related regulation. It has the characteristics of swift compaction, handle resetting automatically and safety protection, the testing result is high precise and reliable. It is the necessary equipment for the paper-making factory, research and quality supervision department.
According to ISO3036 "determination of bursting strength of paperboard" and gb2679.7 "determination of bursting strength of paperboard", the tester is designed with the characteristics of fast pressing, automatic reset of operating handle and reliable safety protection. The tester has a wide range of applications and is an indispensable common instrument for enterprises and departments of paperboard, carton production, scientific research and commodity inspection
Main technical parameters
Digital Type paperboard Puncture Strength Tester
(1) Micro Computer-controlled technology, open structure, high automation procedures, simple and convenient operation, safe and reliable.
(2) Automatic measurement, intelligent judgement, real time display of clamping pressure for sample by operation system.
(3) Automatically measure, count, print the test results, and data can be stored.
(4) Micro-high-speed thermal printer, high-speed print, low noise, no need for ink and ribbons, and it is easy to use and has low failure rate.
|Indication Error||Ruler A: ±0.05J; Ruler B:± 0.1J; Ruler C: ±0.2J; Ruler D: ±0.5J|
|(within the limit range 20%~80% at each measurement)|
|Pendulum Shaft Friction||>120 times|
|and Air Resistance||( Measured by the swing frequency of arm swing)|
|Triangular Pyramid||Side Length: 60×60×60 mm; Height: 25±0.7 mm;|
|edge fillet Radius: 1.5±0.1 mm|
|Outside Dimension (L*W*H)||800*500*750 mm|
|Unit ( optional):||kg-cm, Joule, or designated|
|Impale Cone: height:||1|
|Work Environment||Temperature: 5~35°C;|
|Relative Humidity: ≤85%|
|Power||AC220V ±10%, 2A, 50Hz|
|Standard||GB/T2679.7 , ISO3036 , ASTM D781 , TAPPI T803|
Install the appropriate counterweight weight on the swing arm, hang the swing arm at the starting position, close the locking block, lock the locking screw, put the friction sleeve ring on the back of the piercing head, select the test gear according to the counterweight weight weight, clamp the sample to be tested between the upper and lower splints, open the screw, release the locking block, swing, the piercing head passes through the sample, and when the pendulum swings back Take the swing arm by hand, lift the pendulum, put back the locking block, lock the screw, read the measured value of the display screen, and prepare for the next test.
If the result is less than 12j, it is accurate to 0.1j; if the result is greater than 12j, it is accurate to 0.2j
Contact Person: Andy Xiao