Salt spray tester is to test the anti corrosion capability of protection, painting layer, some spare parts and electronic components.
Generally, there are two kinds of tests of salt spray tester: NSS test and ACSS test. What’s the difference between them?
1. Temperature difference:
Test chamber: 35℃±1℃ Pressure bucket: 47℃±1℃
Test chamber: 50℃±1℃ Pressure bucket: 63℃±1℃
For NSS test, typical PVC board can satisfied this requirement. But for ACSS test, the chamber material needs to resist high temperature and strong acid. PP board can make it.
NSS test: GB/T2423.17-2008, GB/T2423.18-2000 GB/T10125-1997,
GB/T10587- 2006, GB10593.2-1990, GB/T1765-1979, GB/T1771-2007,
ACSS test: All standards above, and with IEC、MIL、DIN、ASTM、IS、CNS standards.
China: NacI Distilled water; Potency: (50±5)g/l; PH 6.5~7.2
USA: NacI Distilled water; Potency:5%±1%; PH 6.5~7.2
Germany:NacI Distilled water; Potency: (50±5)g/l; PH 6.5~7.2
Japan: NacI Distilled water; Potency:5%±1%; PH 6.5~7.2
France: NacI Distilled water; Potency:5%; PH 6.5~7.2
If you have any interest of the products above, or if you need any professional advice of those machines, welcome to contact us.